Bench Scale Sized Freeze Dryer, Vacuum Freeze Dryer Supplier
TOPT-10C laboratory lyophilizer is a type of Multi-pipe Vacuum Freeze Dryer, which is widely used in
medicine, pharmacy, biology research, chemical industry and food production,etc.
1. The vacuum pump and main engine is connected by pumping vacuum pressure pipe. Both ends using
standard quick clamp.Coate amount of vacuum grease on the sealed rings before connecting.
2. Connect the main engine power and vacuum pump power.
3. Please read carefully the vacuum pump manual. Check the vacumm pump and confirm has been filling
the vacuum pump oil. Don't run without oil.
4. The white sealed ring embedded in glass door should be keeped clean. Coate a layer of vacuum grease
5. The style of defrost is electric defrost. Generally, the defrost time is less than 30 minutes.
1. The vacuum can not achieve below 15Pa:
(1) Check the release valve is whether fastened down or not.
(2) The vacuum pump and main engine is connected by vacuum tube. Check the clamp is whether fastened
down or not.
(3) Check the bottom of organic glass cover is whether smooth or not. Check the “O” rubber ring is whether
damaged or not.
(4) Check the vacuum pump oil. Generally, oil should be changed after working 200 hours.
2. The vacuum pump leaks oil:
Check the leaking part (is possible sealed pad). Replace the corresponding parts.
3. the temperature of cold trap is high:
The temperature of condition is high. If the cooing system is break down, please connect with our company
More laboratory vacuum freeze drying machine please contact email@example.com learn more.