How To Use Rotary Evaporator Extraction Method?
Rotary evaporator is an ideal instrument necessary for the concentration, drying and recycling relied on by
the chemical industry, pharmaceutical industry, foods, environmental protection, universities and colleges,
and research laboratories and so on, and especially adaptable to the small scale production unit.
Simple working principle of rotary evaporator: put the rotary evaporation bottle (flask used to increase the
evaporation area) in thermostatic water bath under the condition of reduced pressure, and make the solution
diffuse and evaporate efficiently while being rotated.
Easy operation steps of rotary evaporator evaporation machine:
1.Installation well, and connect power: 220v/50Hz, or 110V.
2.Press power switch to connect it, green light display, rotating speed governing botton, motor will rotating.
3.Press lifting key (▲/▼) , motor head could lifting automatic (up/down).
4.After add water in water bath, press heating switch (on), set temperature then will automactic control heating,
strictly prohibit heating without water.
5.Adjust motor head angle, loose vertical column right plum blossom knob firstly, rotate motor head, after arrive
suitable angle, tight it is ok.
6.After finish operating, power and heating switch need to close (off), pull out power source plug.
As you see, now all rotary evaporator has automatic lifting function, you just need to set the parameters your need,
such as rotating speed, heating temperature, etc, the whole system can be running automatic, this is very convenient
and high efficient for you. Even timer function also optional, so nothing is worth worried, make your research more
accurate & efficient.
Want to know more detail information or want to see the rotary evaporator operation steps video, welcome contact
firstname.lastname@example.org learn more.